The portable soil moisture detector adopts the latest digital integrated circuit technology and international testing technology to design a new type of intelligent handheld tester. It uses a large full-color LCD screen to display readings in real time. At the same time, the test circuit is designed by the digital chip of the international big factory. Very high sensitivity and excellent repeatability can be achieved. The shell is made of industrial high strength composite plastic, good strength. The shape of the sensor is hand-held design, convenient for users to carry, with good corrosion resistance, strength and toughness.
* Model: JXBS-3001
* Minimum order quantity: 1 piece
* Delivery date: within 24 hoursFeatures
1. Simple to use, less operation steps, fast measuring speed, no reagent, no limit on the number of tests.
2. High measuring accuracy, fast response speed and good interchangeability.
3. The electrode is made of specially treated alloy material, which can withstand strong external impact and is not easy to damage.
4. Completely sealed, acid and alkali corrosion resistance, can be buried in the soil for long-term dynamic test.
5. Probe plug-in design, accurate measurement, reliable performance.
Model: JXBS – 3001
Power supply: lithium battery
Moisture measurement range: 0-100%
Moisture measurement accuracy: The reading of plus or minus 3% ~ 5%
Display: Three rows 105*55cm
Storage: 200,000 sets
Protection class: IP68
Power supply: 12V DC, 2500MA.h (standard)
Display mode: 2.8 inch liquid crystal display
Output signal: RS485/ 4-20mA /0-5V/0-10V
Data storage: 4MB storage (standard configuration)
Communication protocol: Modbus serial communication protocol
How to Use a portable soil moisture detector ?
1. Rapid detector
When measuring the position, avoid rocks and ensure that the steel needle does not touch hard objects. Discard the topsoil at the required measurement depth, maintain the original tightness of the underlying soil, and insert the sensor vertically into the soil. Don’t rock from side to side. It is recommended to make several average measurements in a small area of a measuring point.
2. inserted ground measurement method
Dig a pit with a diameter >20cm vertically, insert the induction needle horizontally into the pit wall to a predetermined depth, and fill the pit tightly. After a period of stability, measurements and recordings can continue for days, months, or even longer.
1. How to find the “Data View” menu?
In the “View Data” menu, you can view each data of the record one by one, starting from the record. On the top of the data is the time of the data recording. An interface displays four data, press left or right to view the lower half data, and press
2. How to query sensor data?
When you purchase a sensor with the reporting function, you can view data remotely through the Precision Cloud server for convenience and convenience.
Sensor devices connect to the cloud server through 4G/ NB-iot network module and report data to the cloud server according to the default frequency (frequency can be adjusted).
You only need to log in to the server using the registered Jingxun Cloud account, and use the unique identity ID of each host to query device data.
At the same time, the platform also has the alarm function. You only need to set the alarm value through the platform. When the monitoring data exceeds the alarm value, the platform will notify you through SMS or QQ email, so that you can monitor the data more conveniently and quickly.