This IoT wind speed sensor shell is made of high-quality aluminum alloy profiles, and the exterior is electroplated and sprayed. It has good anti-corrosion and anti-corrosion characteristics, which can ensure the long-term use of the transmitter without corrosion. We offer rs485 wind speed sensor, 4-20mA wind speed sensor, 0-5V wind speed sensor, 0-10V wind speed sensor, pulse wind speed sensor.
- Model: JXBS-3001-FS
- MOQ: 1 PCS
- Delivery date: within 24 hours
IoT wind speed sensor Features
1. Our 360 degree wind direction sensor ranges from 0 to 359 degrees. The wind direction sensor is small and light, easy to carry and assemble.
2. Anti-electromagnetic interference processing. Adopt high performance imported bearings, small rotation resistance, accurate measurement.
3. Polycarbonate shell, high mechanical strength, high hardness, corrosion resistance, no rust, can be used outdoors for a long time.
4. The structure and weight of the equipment are carefully designed and distributed. The moment of inertia is small and the response is sensitive.
5.When we use the wind speed sensor, we can connect the Jingxun cloud to use together. Users can configure the configuration interface according to the field situation, and can view and alarm data at any time.
wind speed sensor Parameters
|Parameter name||Parameter value|
|Response time||≤ 5 seconds|
|Accurate measurement Material||high-quality aluminum alloy|
|Polycarbonate Power consumption||≤ 1W|
|Communication Port||pulse/rs485/4- 20mA/0-5V/0-10V|
Aluminum wind speed is widely used in wind speed measurement in greenhouses, environmental protection, weather stations, ships, docks, and breeding. To facilitate you to view data at any time, we provide a free cloud platform for viewing data on a computer or mobile phone.
1. Installation note:
Flange mounting, threaded flange connection makes the lower pipe fitting of the wind speed sensor firmly fixed on the flange, the chassis ø65 mm, four ø6 mm mounting holes are cut in the Ø47.1mm circumference, and they are tightly bolted to the bracket, to keep the whole instrument in the best level, to ensure the accuracy of wind speed data. Flange connection is easy to use and can withstand greater pressure.
2. Preventive Measures:
Do not disassemble by yourself, not to mention touch the sensor core, so as not to damage the product.
Keep away from high-power interference devices, such as inverters and motors, to avoid inaccurate measurement. Disconnect the power supply before installing or removing the transmitter. The water in the transmitter may cause irreversible changes.
Prevent chemical agents, oil, dust and other direct intrusion into the sensor, do not use in condensation and extreme temperature environment for a long time, strictly prevent thermal shock.
1.What is the service life of JXCT wind speed sensor?
Under normal environment, our wind speed sensor has a service life of 5-8 years, and the overall warranty is 2 years.
2.How to choose a wind speed sensor？
First, consider the use environment and choose the appropriate wind speed sensor according to the environmental requirements. If used with a weather station, you can choose a polycarbonate wind speed sensor. For long-term outdoor use, you can choose an aluminum shell wind speed sensor. If you want to measure the wind speed in the tube, choose the duct-type wind speed sensor, if you want high-precision data, you can use an ultrasonic wind speed and direction sensor.
Connect the anemometer to the device, slowly turn the wind blade by hand, measure the voltage between the two wires of the anemometer with the multimeter voltage file, and you can see that the multimeter voltage reading is between high (such as 5v) and low (such as 0v) under normal conditions. change. This is the result of low-speed pulse measurement.
An anemometer is an instrument that measures air velocity. There are many types of it. The most commonly used wind cup anemometer in meteorological stations is the sensing part of three parabolic cones and empty cups fixed on a support at 120° each other. The concave surfaces of the empty cups all face one direction. The whole sensing part is installed on a vertical rotating shaft. Under the action of wind, the wind cup rotates around the shaft at a speed proportional to the wind speed.